Snap Electrode Ecg Accessories Holter Cable 5 Leads For Patient Use
Holter ECG Cable 5 Leads For Patient Use Snap Electrode
ECG Cable Introduction
1:Series ECG Cable Lead Sets as a spare parts of the
2:It's transmit the pysiological sign from the patient's
body,connecting Electrocardiograph the with the electrode;
3:the appearance of the play is diffenrent, according to the
1:Insulated performance:Between wire and wire ,wire and shield ,the
insulated resistance should >100MΩ ,the testing voltage should
be 100 DC.
2:Strength of anti electricity , between wire and wire ,wire and
shield,exert AC1000V±20%,50Hz voltage,1 s, creepage
current<0.25mA, 1 s, creepage current≤0.25mA
3:Creepage current: use the 120V AC 60HZ power supply, live wire
tadem ammeter and all trunk cable and lead wire signal cable, the
ammeter indicate the passed creepage current less than 10A
4:Anti defibrillation performance(sutible for the product with
defibrillation function) The product suffer DC5000V 360J(joule)20%
power surge 3 times (every time exert electricity power time is
60seconds) Change in resistance value does not exceed ±10%
After Sale Service:
1.One year warranty for the products.
2.Please send back the defective goods which are within the
warranty period, we will repair them or send you new products for free.
3.If the sample or small quantities (the fee is less or around shipping fee) are damaged, please send us the pictures or video to confirmthe problem, then we will send you new goods for free.